I have 2 Davis buckets, one is modded to the OS station and a ClimeMet Manual (100mm funnel)
The OS (Davis modded bucket) also has a larger funnel (250mm) and tips at about 0.1mm
Early hours we had a wee downpour and i checked how much we had and this is what was recorded
There is no getting around the manual, imo, that has to be right?
I have only had the Davis since Feb, surely it shouldnt need calibrating?
Any thoughts please?